The Fluke 717 Pressure Calibrators delivers outstanding performance, durability and reliability. These calibrators are compact, lightweight, and easy to carry. With a push button interface similar to the multifunction Fluke 740 Series Documenting Process Calibrators, the 717s are easy to use. They are EMI tolerant, dust and splash resistant, and feature a removable battery door for quick battery changes.

These calibrators come in 9 models measuring pressure up to 5000 psi/345 bar and are compatible with Fluke 700Pxx differential, gauge and absolute pressure modules to increase functionality.

<img class="leftAlone" title="" src="https://unitestinst.com/assets/Uploads/717.jpg" alt="717" width="120" height="220" />

<strong>Features at a Glance</strong>
<ul class="PrdPlainText">
 	<li>Measure pressure, 0.05% of full scale with internal sensor up to 5000 PSI/345 bar sensor (5000G model)</li>
 	<li>Measure mA with 0.015% accuracy and 0.001 mA resolution, while sourcing 24V loop power</li>
 	<li>Measure pressure to 10,000 psi/700 bar using one of 29 Fluke 700Pxx Pressure Modules</li>
 	<li>1/8 NPT female pressure fitting</li>
 	<li>Compatible with non-corrosive Gases and liquids</li>
 	<li>Select eleven different engineering units</li>
 	<li>Zero, min, max, and hold functions</li>
 	<li>Calibrate P/I instruments with simultaneous pressure and current measurement</li>
 	<li>Pressure switch test function captures set, reset and Deadband values of a pressure switch</li>
 	<li>Error% calculation of mA measurements for quick pass-fail decisions</li>
 	<li>Long lasting battery life</li>
 	<li>Holster features test lead storage</li>
 	<li>Compatible with Toolpack magnetic hanging system</li>
 	<li>New switch test</li>
</ul>