<strong>Unmatched High Accuracy for Indicators, Transmitters or Data Loggers</strong>

The HC2-SH / HC2-S3H fulfills the highest demands on measuring accuracy. It measures humidity and temperature and calculates the dew/frost point.

<strong>Applications</strong>
HVAC, food industry, building services equipment, paper, textile and pharmaceutical industries

<strong>Features</strong>
<ul>
 	<li>Accuracy: ±0.5 %rh, ±0.1 K, at 23 °C ±5 K</li>
 	<li>Range of application: -50…100 °C / 0…100 %rh</li>
 	<li>Digital interface (UART) and scalable analog outputs, 0…1 V</li>
 	<li>Standard output scaling: 0…1 V = -40…60 °C / 0…100 %rh</li>
 	<li>Adjusted at 23 °C and 10, 20, 30, 40, 50, 60, 70, 80, 90 %rh, then calibrated at 20, 50, 80 %rh</li>
</ul>