<strong>The fastest, one-handed tool for loop checks.</strong>

The Fluke 707 Loop Calibrator is a high performance solution for calibration, repair and maintenance of current loops. The new "Quick Click" knob making it extremely fast and easy to use. It features:
<ul class="prdplaintext">
 	<li>One handed user interface with "Quick Click" detented knob</li>
 	<li>mA sourcing, simulation and measurement</li>
 	<li>0.015% accuracy and 0.001 mA resolution</li>
 	<li>Simultaneous mA and % of span display</li>
 	<li>Select able step or ramp outputs</li>
 	<li>24 V Loop supply with mA measure, including 250Ω HART resistor</li>
 	<li>0-28 VDC measurement to check Loop Voltage</li>
 	<li>Easy battery exchange without opening the case</li>
 	<li>Compact case with separate protective holster</li>
 	<li>Three year warranty</li>
</ul>
<img class="leftAlone" title="" src="https://unitestinst.com/assets/Uploads/707-app.jpg" alt="707 app" width="250" height="292" />

<strong>The fast, one-handed tool for loop checks. Just click to calibrate. </strong>
This feature-rich solution combines the best features push-button and knob-base loop calibrator and offers:
<ul class="prdplaintext">
 	<li>Large display and simple, quick-click interface for easy one-handed operation.</li>
 	<li>Simultaneous mA and % readout for quick, easy interpretation of readings.</li>
 	<li>mA accuracy of 0.015%, superior to other loop calibrators in this price range.</li>
 	<li>1 µA resolution for mA source, simulate and measure.</li>
 	<li>Push button 25% steps for fast, easy linearity checks.</li>
 	<li>0-100% “span check” for fast confirmation of zero and span.</li>
 	<li>Selectable slow ramp, fast ramp, and step ramp to provide smooth outputs for valve slewing and loop functional tests.</li>
 	<li>24 volt internal loop supply, so you can power and read a transmitter at the same time without carrying a DMM.</li>
 	<li>Measures V dc to 28 V.</li>
 	<li>Single 9V battery that is easily changed without braking calibration seal</li>
 	<li>0 - 20 mA or 4 - 20 mA default start up modes.</li>
 	<li>Innovative output adjustment dial with 1 µA and 100 resolution.</li>
 	<li>HART ™ mode connects 250 Ω resistor in series with 24 V loop for compatibility with HART communicators.</li>
</ul>
<strong>Safety conformance</strong>
CSA C22.2 No. 1010.1:1992