<strong>Specially designed for transport applications</strong>

<span class="PrdPlainText" data-mce-mark="1">The Fluke 437 Series II 400Hz Power Quality and Energy Analyzer is designed specifically for the defense and avionics industries. Capable of measuring up to 400 Hz, the 437 II is a must-have for those working in submarines, aircrafts, and other transport applications.</span>

<strong><span class="PrdBoldText" style="font-size: medium;" data-mce-mark="1">Applications</span></strong>
<ul class="PrdPlainText">
 	<li><span style="font-size: x-small;"><strong><span class="PrdBoldText" data-mce-mark="1">400 Hz measurement</span></strong></span> – capture power quality measurements for avionic and military power systems.</li>
 	<li><strong><span class="PrdBoldText" data-mce-mark="1">PowerWave data capture</span></strong> – capture fast RMS values to see every waveform so you can determine how the voltage, current and frequency values are interacting.</li>
 	<li><strong><span class="PrdBoldText" data-mce-mark="1">Power inverter efficiency</span></strong> – measure both ac power and dc power, in and out, to monitor the efficiency of inverters.</li>
 	<li><strong><span class="PrdBoldText" data-mce-mark="1">Energy monetization</span></strong> – calculate the fiscal cost of energy waste due to poor power quality.</li>
 	<li><strong><span class="PrdBoldText" data-mce-mark="1">Energy assessment</span></strong>– quantify the before and after installation improvements in energy consumption to justify energy saving devices.</li>
 	<li><strong><span class="PrdBoldText" data-mce-mark="1">Frontline troubleshooting</span></strong> – quickly diagnose problems on-screen to get your operation back online.</li>
 	<li><strong><span class="PrdBoldText" data-mce-mark="1">Predictive maintenance</span> </strong>– detect and prevent power quality issues before they cause downtime.</li>
 	<li><strong><span class="PrdBoldText" data-mce-mark="1">Long-term analysis</span></strong> – uncover hard-to-find or intermittent issues.</li>
 	<li><strong><span class="PrdBoldText" data-mce-mark="1">Load studies</span></strong> – verify electrical system capacity before adding loads.</li>
</ul>
<table style="width: 100%;" border="0" cellspacing="0" cellpadding="0" align="left">
<tbody>
<tr valign="top">
<td valign="top">
<ul>
 	<li>400 Hz measurement: By increasing the power frequency to 400 Hz, transformers and motors can be much smaller and lighter than at 50 or 60 Hz, which is an advantage in aircrafts, submarines, space crafts, and other military equipment and hand-held tools. The 437 II model captures power quality measurements for these types of avionic and military systems.</li>
 	<li>Power inverter efficiency: Simultaneously measure AC output power and DC input power for power electronics systems using optional DC clamp.</li>
 	<li>PowerWave data capture: Capture fast RMS data, show half-cycle and waveforms to characterize electrical system dynamics (generator start-ups, UPS switching etc.).</li>
 	<li>Energy loss calculator: Classic active and reactive power measurements, unbalance and harmonic power, are quantified to pinpoint the fiscal costs of energy losses.</li>
 	<li>Troubleshoot real-time: Analyze the trends using the cursors and zoom tools.</li>
 	<li>Highest safety rating in the industry: 600 V CAT IV/1000 V CAT III rated for use at the service entrance.</li>
 	<li>Measure all three phases and neutral: With included four flexible current probes with enhanced thin flex design to fit into the tightest places.</li>
 	<li>Automatic Trending: Every measurement is always automatically recorded, without any set-up.</li>
 	<li>System-Monitor: Ten power quality parameters on one screen according to EN50160 power quality standard.</li>
 	<li>Logger function: Configure for any test condition with memory for up to 600 parameters at user defined intervals.</li>
 	<li>View graphs and generate reports: With included analysis software.</li>
 	<li>Battery life: Seven hours operating time per charge on Li-ion battery pack.</li>
</ul>
<table border="0" cellspacing="0" cellpadding="0">
<tbody>
<tr>
<td colspan="4"><strong>PowerWave data capture</strong>
<span class="PrdPlainText" data-mce-mark="1"><span class="PrdPlainText" data-mce-mark="1">For some users, loads switching is a cause of power quality problems. When loads switch on, the current draw sometimes causes the voltage to drop to a level that causes other equipment to malfunction. The PowerWave function available in the 435 and 437 Series II models enables users to capture voltage, current and frequency signals simultaneously at a high speed to see which interaction is potentially causing problems. PowerWave goes beyond standard power quality measurements; PowerWave’s fast data capture mode enables system dynamics to be characterized.</span></span>Waveforms for voltage and current are continuously captured for the specified time, and are displayed on screen in high detail; the power waveform is derived from the data. In addition, half-cycle RMS values for voltage, current, power and frequency can be stored and retrieved for analysis. This feature is particularly useful for testing of standby generation systems and UPS systems where reliable switch-on can be vital.
<span class="PrdPlainText" data-mce-mark="1">These UPM calculations are used to quantify the fiscal cost of energy loss caused by power quality issues. The calculations are computed, along with other facility-specific information, by an Energy Loss Calculator that ultimately determines how much money a facility loses due to wasted energy.</span></td>
</tr>
<tr>
<td colspan="4"><strong>Power inverter efficiency</strong>
<span class="PrdPlainText" data-mce-mark="1">Power inverters take DC current and transform it into AC current, or vice versa. Solar generation systems usually include an inverter that takes the DC energy from the solar cells and converts it to useful AC power. Inverters can lose performance over time and need to be checked. By comparing the input power with the output power you can determine the system efficiency. The 435 and 437 II models can measure the efficiency of such inverters by simultaneously measuring the DC and AC power of a system to determine how much power is lost in the conversion process.</span></td>
</tr>
<tr>
<td><img class="leftAlone" title="" src="https://unitestinst.com/assets/Uploads/Energy-Loss.jpg" alt="Energy Loss" width="320" height="240" /></td>
<td><strong>Unified power measurement</strong>
<span class="PrdPlainText" data-mce-mark="1">Previously, only experts could calculate how much energy was wasted due to power quality issues; utilities could calculate the cost, but the required measurement process was beyond the reach of average electricians. With the new, patented Unified Power function of the 430 Series II, you can use one handheld tool to determine how much power is being wasted, and calculate exactly what the extra consumption costs. Fluke’s patented Unified Power Measurement System provides the most comprehensive view of power available, measuring:</span>
<ul>
 	<li>Parameters of Classical Power (Steinmetz 1897) and IEEE 1459-2000</li>
 	<li>Detailed Loss Analysis</li>
 	<li>Unbalance Analysis</li>
</ul>
</td>
</tr>
<tr>
<td colspan="4"><span class="PrdPlainText" data-mce-mark="1">These UPM calculations are used to quantify the fiscal cost of energy loss caused by power quality issues. The calculations are computed, along with other facility-specific information by an Energy Loss Calculator that ultimately determines how much money a facility loses due to wasted energy.</span></td>
</tr>
<tr>
<td><img class="leftAlone" title="" src="https://unitestinst.com/assets/Uploads/V-A-HZ-US.jpg" alt="V A HZ US" width="320" height="240" /></td>
<td><strong>AutoTrend - Quickly see the trend</strong>
<span class="prdplaintext" data-mce-mark="1">Unique AutoTrend gives you fast insight into changes over time. Every displayed reading is automatically and continuously recorded without having to set up threshold levels or interval times, or having to manually start the process. You can quickly view trends in voltage, current, frequency, power, harmonics or flicker on all three phases plus neutral. And you can analyze the trends using the cursors and zoom function – even while background recording continues</span></td>
</tr>
</tbody>
</table>
<strong>SystemMonitor - Check performance against EN50160 with ease
</strong>With a single push of a button, the unique System-Monitor gives you an overview of power system performance, and checks the compliance of incoming power to EN50160 limits or to your own custom specifications. The overview is shown on a single screen, with color-coded bars clearly indicating which parameters fall outside the limits.

<img class="leftAlone" title="" src="https://unitestinst.com/assets/Uploads/435-systemmonitor.gif" alt="435 systemmonitor" width="300" height="193" />

<span class="PrdItal" data-mce-mark="1"><em>The System-Monitor overview screen gives instant insight into whether the voltage, harmonics, flicker, frequency and the number of dips and swells fall outside the set limits. A detailed list is given of all events falling outside the set limits.</em></span>
<table border="0" cellspacing="0">
<tbody>
<tr>
<td><img class="leftAlone" title="" src="https://unitestinst.com/assets/Uploads/435-logger.gif" alt="435 logger" width="175" height="132" /></td>
<td><img class="leftAlone" title="" src="https://unitestinst.com/assets/Uploads/435-power.gif" alt="435 power" width="175" height="132" /></td>
<td><img class="leftAlone" title="" src="https://unitestinst.com/assets/Uploads/435-harmonics.gif" alt="435 harmonics" width="175" height="132" /></td>
</tr>
<tr>
<td><span class="PrdItal" data-mce-mark="1"><em>Logging function allows you to customize measurement selections and provides instantaneous analysis of user-selectable parameters.</em></span></td>
<td><span class="PrdItal" data-mce-mark="1"><em>Measure and record power (W), VA and VARs. The 434 adds the ability to record energy consumption.</em></span></td>
<td><span class="PrdItal" data-mce-mark="1"><em>Track harmonics up to the 50th, and measure and record THD in accordance with IEC61000-4-7 requirements </em></span></td>
</tr>
</tbody>
</table>
</td>
</tr>
</tbody>
</table>