<strong>Class-A compliance for the most demanding power quality tests</strong>

The Fluke 1760 Three-Phase Power Quality Recorder is fully compliant to IEC 61000-4-30 Class-A, for advanced power quality analysis and consistent compliance testing. Designed for analysis of utility and industrial power distribution systems, in medium- and low-voltage networks, this power quality monitor provides the flexibility to customize thresholds, algorithms, and measurement selections. The 1760 power recorder captures the most comprehensive details on user-selected parameters.

<strong><span style="font-size: medium;" data-mce-mark="1">Applications</span></strong>

<strong>Detailed disturbance analysis</strong> – Perform high-speed transient analysis and uncover root cause of equipment malfunction for later mitigation and predictive maintenance. The fast transient option, with its 6000 V measurement range, allows capture of very short impulses such as lightning strikes.

<strong>Class-A quality-of-service compliance</strong> – Validate incoming power quality at the service entrance. Thanks to Class A compliance, the Fluke 1760 allows undisputable verification.

<strong>Event correlation at multiple locations</strong> – Utilizing GPS time synchronization, users can quickly detect where a fault occurred first, either inside or outside the facility.

<strong>Galvanic separation and DC coupling</strong> – Allows complete measurements on dissimilar power systems. For example troubleshoot UPS systems by simultaneously recording the battery voltage and power output.

<strong>Power quality and power load studies</strong> – Assess baseline power quality to validate compatibility with critical systems before installation and verify electrical system capacity before adding loads.<img class="leftAlone" title="" src="https://unitestinst.com/assets/Uploads/1760feat-1.jpg" alt="1760feat 1" width="200" height="223" />
<ul class="PrdPlainText">
 	<li><strong>Fully Class-A compliant:</strong> Conduct tests according to the stringent international IEC 61000-4-30 Class-A standard</li>
 	<li><strong>GPS time synchronization:</strong> Correlate data with events or datasets from other instruments with precision</li>
 	<li><strong>Flexible and fully configurable thresholds and scale factors:</strong>Allows user to pinpoint specific issues by defining the detailed criteria for detection and recording of disturbances.</li>
 	<li><strong>Uninterrupted power supply (40 minutes):</strong> Never miss important events - even record the beginning and end of interruptions and outages to help determine the cause</li>
 	<li><strong>10 MHz, 6000 Vpk waveform capture:</strong> Get a detailed picture of even the shortest event</li>
 	<li><strong>2 GB data memory:</strong> Enables detailed, simultaneous recording of numerous power parameters for long periods of time</li>
 	<li><strong>Comprehensive software included:</strong> Provides trend diagrams for root cause analysis, statistical summaries, report writing and real-time data monitoring in the online mode</li>
 	<li><strong>Plug and play:</strong> Allows quick setup with automatic sensor detection; sensors are instrument powered, eliminating the need for batteries</li>
 	<li><strong>Rugged field design:</strong> Insulated housing and a solid state design with no rotating components enable reliable testing under nearly any conditions</li>
</ul>
<strong>There are four Fluke 1760 models to select from:</strong>
<table style="width: 100%; height: 431px;" border="1" cellspacing="0" cellpadding="2">
<tbody>
<tr>
<td>Product Feature</td>
<td>Fluke 1760 Basic</td>
<td>Fluke 1760TR Basic</td>
<td>Fluke 1760</td>
<td>Fluke 1760TR</td>
</tr>
<tr>
<td class="PrdBoldText">Power quality statistics according to EN50160</td>
<td align="center">●</td>
<td align="center">●</td>
<td align="center">●</td>
<td align="center">●</td>
</tr>
<tr>
<td class="PrdBoldText">Voltage event list (dips, swells and interruptions)</td>
<td align="center">●</td>
<td align="center">●</td>
<td align="center">●</td>
<td align="center">●</td>
</tr>
<tr>
<td colspan="5">Continuous recording of:</td>
</tr>
<tr>
<td>Voltage</td>
<td align="center">●</td>
<td align="center">●</td>
<td align="center">●</td>
<td align="center">●</td>
</tr>
<tr>
<td>Current</td>
<td align="center">●</td>
<td align="center">●</td>
<td align="center">●</td>
<td align="center">●</td>
</tr>
<tr>
<td>Power P, Q, S</td>
<td align="center">●</td>
<td align="center">●</td>
<td align="center">●</td>
<td align="center">●</td>
</tr>
<tr>
<td>Power factor</td>
<td align="center">●</td>
<td align="center">●</td>
<td align="center">●</td>
<td align="center">●</td>
</tr>
<tr>
<td>kWh</td>
<td align="center">●</td>
<td align="center">●</td>
<td align="center">●</td>
<td align="center">●</td>
</tr>
<tr>
<td>Flicker</td>
<td align="center">●</td>
<td align="center">●</td>
<td align="center">●</td>
<td align="center">●</td>
</tr>
<tr>
<td>Unbalance</td>
<td align="center">●</td>
<td align="center">●</td>
<td align="center">●</td>
<td align="center">●</td>
</tr>
<tr>
<td>Frequency</td>
<td align="center">●</td>
<td align="center">●</td>
<td align="center">●</td>
<td align="center">●</td>
</tr>
<tr>
<td>Voltage and current harmonics to the 50th / Interharmonics</td>
<td align="center">●</td>
<td align="center">●</td>
<td align="center">●</td>
<td align="center">●</td>
</tr>
<tr>
<td>THD</td>
<td align="center">●</td>
<td align="center">●</td>
<td align="center">●</td>
<td align="center">●</td>
</tr>
<tr>
<td>Mains signaling</td>
<td align="center">●</td>
<td align="center">●</td>
<td align="center">●</td>
<td align="center">●</td>
</tr>
<tr>
<td class="PrdBoldText">Triggered recordings</td>
<td align="center">●</td>
<td align="center">●</td>
<td align="center">●</td>
<td align="center">●</td>
</tr>
<tr>
<td class="PrdBoldText">Online mode (Oscilloscope, transients and events)</td>
<td align="center">●</td>
<td align="center">●</td>
<td align="center">●</td>
<td align="center">●</td>
</tr>
<tr>
<td class="PrdBoldText">Fast transient analysis up to 10 MHz</td>
<td></td>
<td align="center">●</td>
<td></td>
<td align="center">●</td>
</tr>
<tr>
<td class="PrdBoldText">4, 600V voltage probes</td>
<td></td>
<td></td>
<td align="center">●</td>
<td align="center">●</td>
</tr>
<tr>
<td class="PrdBoldText">4 dual-range flexible current probes (1000 A / 200 A ac)</td>
<td></td>
<td></td>
<td align="center">●</td>
<td align="center">●</td>
</tr>
<tr>
<td class="PrdBoldText">GPS time sync receiver</td>
<td></td>
<td></td>
<td align="center">●</td>
<td align="center">●</td>
</tr>
<tr>
<td class="PrdBoldText">Memory</td>
<td colspan="4" align="center">2 GB Flash memory</td>
</tr>
</tbody>
</table>