<strong>PPC4 - Redefining premium performance in a digital pressure calibrator or portable pressure calibrator</strong>
<ul>
 	<li>Now with ranges up to 14 MPa (2,000 psi)</li>
 	<li>4 ppm control precision now to as low as 1 kPa absolute (0.15 psia)</li>
 	<li>Portable pressure calibrator, available with support in eleven languages</li>
</ul>
<strong>And still delivering:</strong>
<ul>
 	<li>Lowest measurement uncertainty available in a transfer standard</li>
 	<li>50:1 control precision turndown for extreme rangeability</li>
 	<li>Open architecture with up to four internal and/or external reference pressure transducers</li>
 	<li>Real time measurement uncertainty calculation and display</li>
</ul>
<strong>From the cal lab bench top to 24/7 fully automated production testing, there’s a PPC4™ digital pressure calibrator configuration perfect for your pressure transducer calibration application.</strong>
<ul>
 	<li>Choice of new advanced local user interface with color display and point-and-click navigation or economical basic interface</li>
 	<li>New Command Interpreter allows PPC4 as drop-in replacement for controllers in legacy pressure transducer calibration and test systems without costly software changes.</li>
 	<li>New “f class” Q-RPT module for full scale performance at lower cost</li>
</ul>
<strong>The basic and advanced PPC4 includes all the pressure transducer calibration features you expect in a state-of-the-art digital pressure calibrator</strong>

The importance of control precision is increasing with the recent trend for covering multiple pressure ranges with a single calibrator/ controller.

Learn more in "Why control precision matters."

Read it now »
<ul>
 	<li>Portable pressure calibrator, rugged enough for mobile applications and standard shipment without special packaging</li>
 	<li>Advanced user interface with full support in 11 languages</li>
 	<li>Covers the absolute range of 1 kPa (0.15 psi) to 14 MPa (2,000 psi) and gauge equivalent, including very low differential pressure</li>
 	<li>Control precision to ± 4 ppm of Q-PRT span, default ± 0.005% of Autoranged range</li>
 	<li>Three different Q-RPT measurement classes available to fit different performance requirements</li>
 	<li>Measured and delivered pressure uncertainties calculated real-time and available on remote and local interfaces</li>
 	<li>Q-RPTs can be located in an external RPM4 so that PPC4 does not need to be shipped or removed for recalibration</li>
 	<li>AutoRange feature optimizes and sets measurement, control and safety features for the specific range of the test beign run with a few simple entries</li>
 	<li>Dynamic and static control modes with default or user specified parameters</li>
 	<li>Common remote command protocol with Fluke Calibration PPC3 pressure controllers</li>
 	<li>Front panel USB connection and free “cockpit software for full PC-based plug-and-play functionality</li>
 	<li>Multiple remote interface options: RS232 and front panel USB included, IEEE-488.2 or Ethernet and rear USB optional</li>
 	<li>Instantaneous switching between absolute, gauge and bidirectional gauge modes without added hardware or calibration requirements</li>
 	<li>Instantaneous switching between control and measure modes with no pressure discontinuity</li>
 	<li>Automated AutoZeroing while vented at atmospheric pressure</li>
 	<li>16 SI and US pressure units as well as user definable uits</li>
 	<li>Simple, objective pressure “ready/not ready indicator with user adjustable criteria</li>
 	<li>Automatic pressure head correction</li>
 	<li>On-board, programmable calibration sequences with UUT tolerance testing</li>
 	<li>Automated leak testing routines</li>
 	<li>Valve drivers option for system design</li>
 	<li>Automated self purging liquid trap (SPLT) accessory available for protection against liquid contamination</li>
 	<li>Flash memory for simple and free embedded software upgrades from flukecal.com</li>
 	<li>…and much more</li>
</ul>