The 2700G Reference Pressure Gauges provide best-in-class measurement performance in a rugged, easy-to-use, economical package. Improved measurement accuracy allows it to be used for a wide variety of applications. It is ideal for calibrating pressure measurement devices such as pressure gauges, transmitters, transducers, and switches. In addition, it can be used as a check standard or to provide process measurements with data logging.
<ul>
 	<li>Precision pressure measurement from 100 kPa (15 psi, 1 bar) to 70 MPa (10 000 psi, 700 bar)</li>
 	<li>Accuracy to 0.02% of full scale</li>
 	<li>Easy-to-use, rugged construction for reliable performance</li>
 	<li>Combine with the 700PTPK or 700HTPK pump kits for a complete portable pressure testing solution for up to 600 psi with the PTP-1 pneumatic pump and up to 10 000 psi with the HTP-2 hydraulic pump</li>
 	<li>Combine with the P5510, P5513, P5514, or P5515 Comparison Test Pumps for a complete bench top pressure calibration solution</li>
 	<li>Test port is 1/4 NPT Male. 1/4 BSP and M20 X 1.5 adaptors are included standard</li>
 	<li>USB communications cable and universal power supply provided standard</li>
 	<li>Can be used with COMPASS® for Pressure software for automated recording of pressure calibration data</li>
</ul>
The 2700G has an easy-to-use interface that allows you to configure the sampling rate, tare value, damping, and auto-off time interval as well as reset the min/max pressure. You can also view the remaining battery life. Select from 21 different standard engineering units, including bar, in H2O, kPa, MPa, and psi. In addition, the supervisor menu can be used to set one custom engineering unit. To ensure against incorrect measurements, each of the above functions can be locked from use through the supervisor menu.

The 2700G Reference Pressure Gauge can be combined with the Fluke Calibration P5500 series of comparison test pumps to make a complete pressure calibration system. The unique test port design of the P5500 series allows for hand tight connection of the 2700G without the use of PTFE tape.

<strong> Use the 2700G with these comparison test pumps for a complete bench top pressure calibration solution</strong>

<strong>P5510 Pneumatic Comparison Pump</strong>
<ul>
 	<li>Pressure range: 0 to 2 MPa (300 psi)</li>
 	<li>Vacuum range: 0 to 80 kPa (24 inHg)</li>
 	<li>2700G and device under test (DUT) can be connected without the use of tools or PTFE tape</li>
 	<li>Built-in hand pump as pressure/vacuum source</li>
 	<li>High-quality needle valve for fine control</li>
</ul>
<strong>P5513 Pneumatic Comparitor</strong>
<ul>
 	<li>Pressure range: 0 to 20 MPa (3000 psi)</li>
 	<li>High pressure pneumatic operation</li>
 	<li>Screw press for fine pressure adjustment</li>
 	<li>High quality needle valve for metering in supply pressure 2700G and DUT can be connected without the use of tools or PTFE tape</li>
</ul>
<strong>P5514 Hydraulic Comparison Pump</strong>
<ul>
 	<li>Pressure range: 0 to 70 MPa (10 000 psi)</li>
 	<li>Bench mount design</li>
 	<li>Screw pressure for pressure generation</li>
 	<li>2700G and DUT can be connected without the use of tools or PTFE tape</li>
 	<li>Operates with a wide range of fluids (Viton seals standard, EP seals available)</li>
</ul>
<strong>P5515 Hydraulic Comparison Pump</strong>
<ul>
 	<li>Pressure range: 0 to 140 MPa (20 000 psi)</li>
 	<li>Built-in priming pump for large volume applications and system priming</li>
 	<li>High-quality screw press for pressure generation and fine control</li>
 	<li>Clear acrylic reservoir for visibility of fluid level and quality</li>
 	<li>Built-in drain plug for easy removal of fluid</li>
 	<li>Sturdy carrying case with lid</li>
 	<li>Operational with a wide range of fluids (Viton seals standard, EP seals available)</li>
 	<li>2700G and DUT can be connected without the use of tools or PTFE tape</li>
</ul>