<strong><span class="BlackSubHead">Versatile and rugged tools for applications with high currents</span></strong>

<span class="PrdPlainText">Confidently take reliable readings with the True-rms, Fluke 355 and 353 digital clamp meters; the tools of choice for high current clamp-on amp meter measurement up to 2000 A. The extra-wide jaw easily clamps around large conductors, typically found in high-current applications. The extra rugged design and CAT IV 600 V, CAT III 1000 V ratings add an extra element of user protection when taking high-powered measurements.</span>

Accurate peak measurements can be taken using the in-rush current mode, ideal for motors and inductive loads. The 355 also measures voltage and resistance, making this an ideal clamp meter for utilities, electrical contractors and industrial service technicians.

<img class="leftAlone" title="" src="https://unitestinst.com/assets/Uploads/F-355-06a-200.jpg" alt="F 355 06a 200" width="200" height="256" />
<ul class="PrdPlainText">
 	<li>Reliably handle a wide range of high-current applications with 2000 A ac + dc true-rms, 1400 A ac, and 2000 A dc</li>
 	<li>The large 58 mm (2.3 in) jaw capacity is suitable for large or multiple conductors</li>
 	<li>CAT IV 600 V, CAT III 1000 V rating for added user protection</li>
 	<li>In-rush current measurement captures ‘power-on’ surge current with accuracy and repeatability</li>
 	<li>High voltage measurement of 1000 V ac + dc true-rms, 600 V ac, and 1000 V dc allows the user to perform multiple tests with only one tool (355 only)</li>
 	<li>Resistance to 400K ohms coupled with a continuity beeper provide the convenience of a multimeter in a clamp meter. (355 only)</li>
 	<li>Accurately measure frequency up to 1 kHz for optimum troubleshooting</li>
 	<li>Quickly analyze readings using the MIN, MAX, and AVG functions</li>
 	<li>A large backlit display allows for easy visibility in low-lit areas</li>
 	<li>Use the display hold feature to capture readings even when the display cannot be viewed</li>
 	<li>Use the low-pass filter to smooth out noisy loads and stabilize readings</li>
</ul>